MEDTRONIC IPRO2 DOCK DRIVER DOWNLOAD
Finally my PC got up to speed! BG meter readings with which to calibrate the sensor values. Enter text from picture: Do not connect more than one Dock to the computer at one time. This should be done a minimum of eight hours before your first iPro2 patient setup.
|Date Added:||22 October 2011|
|File Size:||6.67 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Moving a patient study If you accidentally upload an iPro2 study to the wrong patient record in CareLink iPro, you can move the study to the correct patient record. Warranty Dock, subject to the conditions and exclusions stated docj.
Do not use IV Prep prior to sensor insertion.
If you find blood on top of the sensor adhesive, do not connect the iPro2. The Dock is connected to power. The iPro2 is not suitable for use in the presence of a flammable anaesthetic mixture with air or with oxygen or nitrous oxide. BG meter readings 47 areas to insert sensor 29 exporting data 55 assistance 8 meters supported 6 attaching iPro2 to sensor 35 ongoing use, administrator tasks 59 AUC 53 ongoing use, user tasks 57 opening patient records 42 patient report settings The customer or users of the Deleting User Accounts Select the user account and click the Open user button.
Make sure that the sensor insertion site is not bleeding before connection. Fast and simple process: You may opt out at any time ipfo2 i texting “STOP” in return to a text; ii sending an email to takecontrol medtronic. Contact the 24 Hour HelpLine. Assess results quickly Simple visuals Most important metrics — all on one page 1.
Dispose of the iPro2 according to the local regulations for battery disposal non-incineration.
If the calendar day does not meet this threshold, it will be designated with an X Use Clinical Judgment. Keep track lpro2 cleaning plug uses and discard the cleaning plug after 30 uses.
By pressing ‘print’ button you will print only current page. If the measured field strength in the location in which the iPro2 CGM system is used exceeds the application RF compliance level above, the iPro2 CGM system should be observed to verify normal operation. If the sensor is pulled out by more than a millimeter, the iPro2 will stop collecting data until the sensor is pushed back in place.
Replace the cleaning plug after 30 uses to maintain a water-tight seal. For the specific regulations and test results for your area, please contact your local ipr2o. The iPro2 takes one hour to start up a sensor. This authorization will expire three 3 years from the later of the date you submit this authorization online; or ii the last date products or services were received from Medtronic, but may expire sooner if you revoke it by following the instructions below.
Medtronic Ipro2 Dock Driver Download
The Dock can be connected to the computer or to the wall- powered adapter, which lets you use a regular power outlet for charging. Existing study reports will not be updated automatically based on the new settings. Before you begin When the patient returns after wearing the iPro2, you will need the following: The sensor insertion process takes a few minutes and the patient only needs to be trained in how to record activities in a log book.
Airtime, message and data rates may apply. Offer your patients a life with fewer injections.
Medtronic iPro2 Dock Drivers
To reduce the likelihood of data communication errors, you should relocate either the wireless device or the iPro2 system devices. Bleeding can occur if you insert the sensor at an angle less than 45 degrees. Precautions Make sure that the sensor insertion site is not bleeding before connection. Medtronic Diabetes Global Headquarters Tel: Administrator Tasks If you have administrative privileges, you can modify a user account.
Please register, and get one.