AGILENT 53132A LABVIEW DRIVER
I finally found the manual and tried some I’m trying to make a Labview program for the Agilent NA load box to regulate the voltage and drive a current from the load. We have recently added a reset step to our tests that use the a counter and every so often the counter returns a measurement interpolator calibration failed error during the subsequent setup of the counter. CORR4 and receive answer with correction table 4 1. Sorry I did not understand your reply fully. I have an old Fluke that is able to make this measurement BUT it is not gapless. Looking at the x driver, you will see i
|Date Added:||23 July 2011|
|File Size:||48.9 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
I use a A frequency counter to measure frequency of a signal. I got a commands, but commands are supports GPIB port.
Labview and Agilent frequency counter A – Discussion Forums – National Instruments
I’m trying to make a Labview program for the Agilent NA load box to regulate the voltage and drive a current from the load. Message 4 of 5. Herte is the original c High Stability Time Base.
Hi faheemkhan, Is still an issue for you?
Agilent 5313X Series Counter/Timer
We have 5 systems all with one of these instruments, and it is the only instrument giving us oabview problems. I managed to solve the issue. Hi, Installing the Command Expert v. Actualyl in the default program, which I donwloaded from NI website, there was some unnecessary commands which were making whole devices slowe.
We are occasionally having problems communicating with our new A’s. Can I save data in the device itself and fetch that after every 2 or 3 seconds.
Hi, I can’t seem to figure out configure the A for gap-free, continuous, ongoing measurement. I have an old Fluke that is able to make this measurement BUT it is not gapless. Type to filter by text Filter by tag Sort Sort by date created: I keep getting a time out error and cant get a readout from my vi. Is the commands are same for Because this device is from a 3rd party, you must contact the producer of the device with this type of question.
Agilent X Series Counter/Timer – IEEE (GPIB) Driver for LabVIEW – National Instruments
Message 5 of 5. The examples I’ve found seem to be for an old I want to communicate with the A fr counter via RS Is 5332a possible to measure the time between two pulses that are about 1 second e.
General Use Programming and Remote Communications clear selected. I’m using labview 8.
Agilent Technologies / Keysight Technologies hp5313xa Counter/Timer
Type to filter by text Filter by tag Sort Sort by date agillent Looking at the x driver, you will see i Message 3 of 5. Hello, i just received my new A, and i installed the updated version CORR4 and receive answer with correction table 4 1. Hi, I’m very green in this area, so please excuse my dumb question.